Meade's 178 ED 7" refractor
Meade's Massive APO
The 178ED 7" Refractor
Updated Feb 16, 1998
Updated Sept 17, 1998
Updated Feb 9, 2002
(Note: In the spring of 2002, Meade announced an improved lens cell for this model.
I am waiting for reports from the field on this, but in my opinion, a better lens
cell would go a long way towards mitigating -and even eliminating- the optical
issues described below. Thanks to Meade for responding to this issue.)
(7" f/9 Apochromatic refractor, LX750 equatorial mount, drives, #1697 computer,
8X50 finder, 26 mm Plossl, 2" visual back & diagonal, $6790 as tested)
"Well, what do you see?" It was me, hopping up and down to keep warm.
"Tell you in a minute," said Mike. I could see him racking the focuser back and forth.
We were playing with Mike's new Meade 7" APO refractor. The scope is huge - you
knew that - but the pictures, here and in Meade's catalog, don't do it justice. The
thing is just massive. It's so large that Mike cannot store it, assembled, in his
house. With the tripod collapsed, the cradle still comes perilously close to the ceiling.
"Well?" I was getting impatient.
"Here, you look."
I sidled up to the eyepiece, one hand on the focuser, and looked. The star was Saiph
in Orion. In and out the focuser went. In and out.
And I can tell you what I saw...
But, we're getting ahead of ourselves. The full story of this particular Meade 178ED
APO goes back almost five months, to a warm early-fall afternoon in 1997, when the
scope was first purchased.
Chapter One: The Scope Strikes Out
This sample was purchased new by a friend of mine (we should all have such friends!)
from a local dealer. He went in to buy a Firstscope 80, and walked out with the Meade.
Doesn't it always work that way? The scope was well packed in six different cartons.
According to Mike, the instructions were clear and well thought out. Getting out to
the field with a scope of this size can present problems, as you can imagine. However,
with a small to mid-size car and some imaginative loading, the 178 ED fits, and still
qualifies as a "portable" instrument.
Setup can be accomplished by one person. None of the steps in the setup are particularly
complicated; there's just lots of 'em. Meade has done a good job of breaking down
the assembly into bite-sized pieces. Briefly, here's how the scope gets set up:
1) Open up the legs of the tripod on level ground.
2) Set the equatorial head on top of the tripod base and screw down tightly.
3) One by one, extend the legs of the tripod (you have to do this carefully).
4) Attach the counterweight (and the declination shaft, if you took it off in transport.)
5) Attach the cradle to the equatorial head. Rotate the cradle to horizontal.
6) Place the tube in the cradle and tighten down the wing nuts. This is hard to do if
you're shorter than about 5' 8". Remove the dust cap.
7) Attach the dew shield with one bolt. Attach an additional dew shield if you use one.
8) Attach the finder (it's a hassle, but leaving the finder attached to the OTA is asking
9) Plug in your diagonal and attach all your electrical cables.
Set up can be done in about twenty minutes, if you take your time and do it right. I
should point out that in the dead of winter, the twenty minutes it takes to do all this
stuff in reverse at the end of your observing session can seem like an eternity.
Mike went observing about 6 times with the scope in September and October of 1997. He
complained about uneven "blooming" of the airy disk, and fan-shaped stars. Planets
refused to focus down correctly. He called Meade, and they sent him a kit to collimate
the main lens. It turns out the "kit" was a piece of paper and an allen wrench. Mike
already knew how to square-on an objective lens, but went ahead and did it anyway.
It didn't help. Meade agreed to take the OTA back in mid-November, 1997. Mike sent
it back, paying for the shipping and insurance himself. About a week later, he
followed up on the package. No, Meade had no record of ever receiving a 7" OTA from
You can imagine what this did to his state of mind. He called up the dealer, in a
near-panic, and asked for help. The dealer made some calls, and within the hour,
they received word that the OTA had been found -- it was sitting on the shipping dock.
Chapter Two: The Scope Returns
Another 5 weeks passed.
One day, Mike got word that the OTA had been returned to the dealer. I went with him
to get it, first thing on a Saturday morning. There was no note, no repair slip, no
paperwork at all. We agreed to meet that night at our usual observing spot.
I showed up at dusk, and brought along my Takahashi for comparison (that's the "little"
scope at the left of the first photo). I set up the Tak on the Super Polaris, then helped
Mike with some lifting on the big Meade. Then we waited.
The 178ED didn't seem to want to settle down. The FS102 was already delivering its
usual knock-your-socks-off images on Saturn, so we enjoyed this for while.
After an hour, the Meade seemed to come to equilibrium. Here is what I saw:
When you defocus a star, you expect the disc of light to expand evenly around the
central airy disc. This was not the case here. Rather, the disc of light seemed to
grow out towards 7 o'clock. Fully defocused, the airy disc actually sits on the outer
diffraction ring - very strange.
Also, we could not bring Saturn to focus. The image was always a little fuzzy.
Frustrated, we retired back to Mike's house, where he showed me the letter he had
written, originally describing the problem to Meade in November, 1997. His description
matched what I saw almost exactly. As a result, we both came to the conclusion that
Meade had not done a thing to the OTA. Perhaps they had meant to, but somehow the
scope just got sent back. Keep in mind, Meade had kept it for almost 2 months.
Once again, arrangements were made to send the scope back, this time at Meade's
expense. Three more weeks passed, and Mike again got word that the scope had
returned to the dealer. Mike called Meade for a detailed explanation of what
they'd done to it, but all he could get was "we collimated it," which could mean
any number of things.
Chapter Three: And the Verdict is...
"OK, Ed, what do you think?" Now it was his turn to hop back and forth, trying to
Mike had taken no chances this time. Instead of going to our usual spot, he set up
the Big Guy in his backyard. Sure, there was some skyglow, but he wanted his tools,
spare eyepieces, etc within easy reach.
Here's what I saw:
The scope shows very little spherical aberration. The diffraction rings looked very
pleasing. However, the airy disc is still a little fan-shaped, and seems to shift just
a bit when you de-focus. The scope was clearly better than before. I would estimate
that they cured 90% of the problem. False centering on the objective? "V" spacing
between the 2 elements? Any ideas from the readers out there?
I also noticed a moderately prominent purple halo around bright objects at mid-to-high
power. How Meade gets away with calling this an "apochromat" is beyond me.
Saturn was low in the sky and didn't look very good. It was hard to tell whether the
scope or the conditions were responsible, so we suspended judgement. Castor, Beta
Monoceros, and Alnitak were split, but not cleanly. This was disturbing. Airy discs
refused to remain points at high power.
However, in a strange turn of events, the Meade turns out to be a superb deep-sky
telescope. Even from this suburban location, the 178ED delivered perhaps the finest
view of the Orion Nebula I've ever seen. Six stars were visible with direct vision
in the Trapezium, and the nebulosity showed fine shadings I'd never noticed before.
NGC2158, the little cluster next to M35, shows some granularity. Perhaps it could
be resolved under darker skies. And I spent an enjoyable 20 minutes or so looking
at M81 and M82.
The purpose of a review, in my opinion, is to reach a sort of conclusion to help
a prospective buyer decide whether a particular product is right for them. In this
manner, I am somewhat stymied for a concrete opinion.
This particular 178ED APO is a wonderful deep-sky instrument that could keep a
galaxy hunter happy for years. However, planetary and double star fans might have
a different opinion. Then there is the issue of Meade's service. Mike reports that
dealing with Meade's support was like fighting an uphill battle.
Myself? I like the scope in its present condition. I like it a lot. Would I pay
$6790 for it? Well... In the end, each user will need to decide the issue of
perceived value (or lack thereof) for themselves.
And that is perhaps the best conclusion I can reach.
Postscript: A New Hope?
Discouraged by the performance of his 178ED, especially in continued comparisons
to the FS102 and the A-P Star 12, Mike contacted Meade again and had a long talk
with them about the situation. In short, he was still not satisfied with the telescope.
When you pay almost $7000 for a refractor, you expect it will provide excellent
views of the planets and double stars. This one doesn't. Meade agreed, and made
arrangements to take the OTA back for a third time.
This time they kept the scope for a long time, from April through mid-September of
1998. During this period there were a number of conversations between Mike and
the factory. Meade assured him that they were taking the situation seriously. The
opticians, he was told had rejected "a half dozen or so" lenses for Mike, and were
looking for an outstanding sample for him. As a result of these conversations alone,
Meade's customer service seemed to improve dramatically. However, this was a
modest consolation at best, since Meade had kept the scope, on and off, for almost
10 months now.
In early September, Meade sent the OTA back to Mike. In conversations with him,
the message given by Meade was something to the effect of, "This is our best, we
cannot do any better than this one." Still, to say that he was apprehensive would
be an understatement. Mike was visibly nervous all week and did not look forward
to having another conversation with the factory should this OTA prove to be as
unsatisfactory as the others (I am sure that Meade had a similar outlook at this
After spending a night alone with the new 178ED, he invited me to take a look through
the scope. And here is what I saw.
A quick inspection of the glass is revealing. The coatings seemed much better than
before. The lenses were so dark I could barely see a reflection of myself, even with
a lamp right behind me. The lens was slightly mis-collimated, but this was easily
Seeing the 178ED set up, I am once again struck by how beautiful the scope is. It
cuts a striking pose against the setting sun. Whoever at Meade was responsible
for the scope's aesthetics really outdid themselves this time.
And what of the optics? They're good. Very good, even. Gone were the coma-
like flares and the off-center airy disks. Stars appeared like tiny round white
balls at higher power. There appeared to be less chromatic aberration in this
sample, although looking at Vega at 165X (using the Meade 9.7 mm Plossl)
does reveal a dim ring of purple around the star.
We split many of the double stars available, including Epsilon Bootes, Pi Aquila,
and the double-double. No problems there. On deep sky, the 178ED showed the
NGC galaxy next to M13. M11 had an almost 3-D quality to it. Stars were sharp
and focused down to pinpoints against black velvet -- exactly the way you'd expect
a quality refractor to perform.
In times of steady seeing, Jupiter revealed considerable detail. The color in
Io was apparent, and detail in the cloud bands was impressive. The 178ED's
1600 mm focal length translates into a generous image scale - most of us weren't
used to seeing the planet so large with mid-range (10-17 mm) Plossls. This
combination of good optics and large image scale alone makes the scope worth
keeping. Planetary observers are going to love this telescope.
All of us present, however thought we had seen these familiar objects look a tad
sharper, or more pleasing in our AP and Takahashi refractors. Part of this is
due, no doubt, to the 178ED's aperture, which is just large enough to be finicky
about seeing conditions.
There were three pairs of eyes present, and some philosophical issues were raised.
How much better is the scope, compared to my 10" Starfinder Dob, which retails
for less than 1/10th the price?
Interesting question. On deep sky, the two scopes' views are more similar than
they are different. On M13, the dob was seeing only about 1/2 magnitude deeper,
although coma and other aberrations were visible. It was a bit surprising that
the refractor could keep up, in terms of light grasp. This leads me to believe
that my dob has issues with light throughput, reflectivity of its mirrors, or
baffling. On closer inspection, the refractor had better contrast, and sharper
stars. This is, of course, part of what you are paying for.
On planets, the dob falls down in comaprison to the 178ED. There's too much
glare, diffraction from the spider gets annoying, and planetary detail never
rises above the fair-to-good range. Finally, the dob obviously lacks the
high-tech gadgetry that accompanies the big refractor.
OK, so the refractor isn't 10X better. But you might as well extrapolate that
to any high end product -- not just telescopes. An FS102/EM2 isn't 10X better
than a C102HD. A Mercedes isn't 10X better than a Geo. Each person will
need to decide where the point of diminishing returns falls off for them. I think
the Meade is still "on the curve," so to speak.
Is the Meade better than the AP130EDT, enough to justify the extra mass and
Good question. The AP has superior optics all around. It has no spherical or
chromatic aberration, both of which we found in the 178ED. But the AP's image
scale is smaller, and it gathers less light. I think most AP130 owners are
going to be spending their time viewing planets or double stars. With the
Meade, you have the option of doing serious deep-sky observing.
There aren't many 7" refractors around, and the asking price for the 178ED,
while high, is reasonable considering what you get. I expect it would not
fare well A/B'ed against an AP180, but the Astro-Physics units aren't
exactly lying about for comparison (they're also NLA.) So the Meade does
fill a niche for somewhat well-to-do observers who want a big refractor.
Is it worth keeping?
Yes. There have been times during the past 10 months when we have said to
each other, "I wonder how this would look in the big refractor?" Through all the
frustrations, I have to say if the scope had been mine, the wait would have been
worth it. There are times when you want some aperture to do deep sky, and there
are times you want a good refractor to look at the planets. While the 178ED
isn't the last word for either application, it's close enough to an all-around
Proud Flagship or White Elephant? Meade's 178ED APO
I think the 178ED is going to be a great school star-party scope. It's
going to impress the heck out of the kids, and they can't hurt it (the
178ED is one of the few scopes I know that will inflict, rather than absorb
damage when bumped into in the dark.)
Complaints were minor. The focuser on this unit wasn't as smooth as on the
other OTAs. There was some focus shift, as well, which was surprising on a
$7000 telescope. The motors are quiet while tracking but make a somewhat ugly
noise when slewing (it's not as bad as an LX200, but the motors do make their
presence known.) Finally, the little levers that lock RA and Dec are identical
(and too close) to the levers that retract the declination shaft and loosen
the counterweight. Anyone in the mood for an accident?
Although we are now more than satisfied with this scope, the situation begs certain
questions. Why wasn't this done right the first time? A certain amount of sample
to sample variation is to be expected from any manufacturer, but we are talking about
a $7000 telescope here. Why did it take nearly 10 months to resolve the problem?
Perhaps most disturbing of all, what did Meade do with the "half dozen or so"
pieces of glass rejected by the opticians this third time around for Mike's OTA?
Is it possible they wound up in other OTAs, with Meade hoping that their new
owners would have neither the tenacity to complain, nor the experience to recog-
nize the problem?
The Meade 178ED, in its present condition, offers good optics for a high, but still
reasonable, price. However, based on our experiences, I would strongly recommend
that prospective buyers look through the one they intend to purchase, to see that
it satisfies them. If this is not possible, be sure and work with a dealer who has
a generous return policy.
Final Update, 2/9/02: The End
During a test at my place comparing a Takahashi to an AP, one of the metal baffles
in the OTA broke loose and lodged itself near the focuser. Once more the scope was
sent back to Meade. A few weeks later it returned. By this time Mike noticed that
the scope was having significant trouble retaining its collimation; even the act of
driving it to a remote location was enough to knock it out of whack. Readers have
told me that Meade is offering a new lens cell that is said to hold the lenses more
firmly. The cost is reasonable, about $100-$200 plus shipping.
After contemplating yet another trip for the scope back to Meade (this would have
been the fifth trip back to the factory in 4 years) Mike finally decided it was time
to move on. A buyer was found, and Mike spent hours on the phone with the potential
purchaser, explaining everything he knew about the scope. He wanted the buyer to
be absolutely sure he knew what he was getting into. The scope left NH for the last
time in February of 2002.
(Note: See also reviews of the 4" 102ED and the 6" 152ED models in this series)
End Review of Meade 178ED APO
Back to Home Page